The behavior of an adaptive optics (AO) system for ground-based high contrast imaging (HCI) dictates the achievable contrast of the instrument. In conditions where the coherence time of the atmosphere is short compared to the speed of the AO system, the servo-lag error can become the dominant error term of the AO system. While the AO system measures the wavefront error and subsequently applies a correction (typically taking a total of one or a few milliseconds), the atmospheric turbulence above the telescope has changed resulting in the servolag error. In addition to reducing the Strehl ratio, the servo-lag error causes a build-up of speckles along the direction of the dominant wind vector in the coronagraphic image, severely limiting the contrast at small angular separations. One strategy to mitigate this problem is to predict the evolution of the turbulence over the delay time. Our predictive wavefront control algorithm minimizes, in a mean square sense, the wavefront error over the delay and has been implemented on the Keck II AO bench. In this paper, we report on the latest results of our algorithm and discuss updates to the algorithm itself. We explore how to tune various filter parameters based on both daytime laboratory tests and on-sky tests. We show a reduction in residual-mean-square wavefront error for the predictor compared to the leaky integrator (the standard controller for Keck) implemented on Keck for three separate nights. Finally, we present contrast improvements for daytime and on-sky tests for the first time.Using the L-band vortex coronagraph for Keck's NIRC2 instrument, we find a contrast gain of up to 2 at a separation of 3 λ/D and up to 3 for larger separations (3-7λ/D).
scite is a Brooklyn-based organization that helps researchers better discover and understand research articles through Smart Citations–citations that display the context of the citation and describe whether the article provides supporting or contrasting evidence. scite is used by students and researchers from around the world and is funded in part by the National Science Foundation and the National Institute on Drug Abuse of the National Institutes of Health.
334 Leonard St
Brooklyn, NY 11211
Copyright © 2023 scite Inc. All rights reserved.
Made with 💙 for researchers
Part of the Research Solutions Family.